In this instrument a considerable linear relative movement of the divided lens corresponds with a comparatively small separation of the double image, so that simple verniers reading to 6 1 0 in.
Measures 1 The circles by Reichenbach, then almost exclusively used in Germany, were read by verniers only.
The word is used also to designate the supporting frame or arms carrying the microscopes or verniers of a graduated circle.
Repsold introduced essential improvements in the meridian circles by substituting microscopes (on Jesse Ramsden's plan) for the verniers to read the circles, and by making the various parts perfectly symmetrical.
The two verniers on the azimuth circle having been read, the magnet is then inverted, i.e.
The mean of all the readings of the verniers gives the reading on the azimuth circle corresponding to the magnetic meridian.
Hence if the readings of the verniers on the azimuth circle are made when the transit is observed we can deduce the reading corresponding to the geographical meridian.
From the mirror magnet and to the east of the latter, and the whole instrument is turned till the centre division of the scale B coincides with the cross-wire of the telescope, when the readings of the verniers on the azimuth circle are noted.
The verniers having been read, the cross-arm is rotated so as to deflect the needle a in the opposite direction, and a new setting is taken.
A collar is provided, which when tightened on the vertical axis, otherwise free to move, holds it rigidly in position with respect to the plate PP. To this collar is attached a slow-motion screw, working against a reaction spring, by which the plate rr can be rotated through a small arc. The upper plate carrying two, three .or four verniers vv is attached to a vertical coned pillar passing through the centre of the larger pillar and rotating in it; this plate can be clamped to the lower plate by means of the screw C, and can be rotated with respect to it by the slow-motion screw d.
The verniers are attached to arms uu bearing on an enlargement of one trunnion of the telescope, one arm pro j ecting downwards and embracing a projection on the standard t.
On the side of the telescope opposite to the horizontal axis is attached a graduated circle g, and, turning concentrically with this circle, is a framework h, to which the readers and verniers of the circle are fixed.
In practice the vertical circle is adjusted once for all, so that when the levels k and l are in the centre of their run, the verniers read true zenith distances.
Assuming, for example, that the northern star has the smaller right ascension, the instrument is first, with the aid of the stop, placed in the meridian towards the north; the verniers of the graduated circle g are set to read to the reading 40-2(Sn+Ss) where 0 is the approximate latitude of the place and Sn, Ss the declinations of the northern and southern star respectively; then the level frame h is turned till the levels k and I are in the middle of their run, and there clamped by the screw m, aided in the final adjustment by the adjoining slow motion screw shown in the figure.