The eyepiece ab consists of two plano-convex lenses a, b, of nearly the same focal length, and with the two convex sides facing each other.
This is known as Ramsden's eyepiece, having been made originally by him.
The eyepiece slides into the tube cd, which screws into the brass ring ef, through two openings in which the oblong frame, containing the micrometer slides, passes.
S is the head of the micrometer screw proper, s that of the screw moving the slide to which the so-called " fixed web " is attached, s' that of a screw which moves the eyepiece E.
5) the eyepiece can be made to follow the star for a considerable distance along a position-web parallel to the screw, the bisection of the web by a star moving by the diurnal motion at right angles to the micrometer screw can only be followed for a limited distance, viz.
8), s that of the screw which moves the eyepiece slide.
One drawback to this form of instrument is that the two webs cannot be viewed simultaneously, and therefore the observer must rely on the steadiness of rate of the clockwork and uniformity in the conditions of refraction whilst the eye is moved from one eyepiece to the other.
Eyepiece sliding in a tube B, as (fig.
253) on a like subject, proceed to describe a most ingenious form of " Ghost Micrometer," in which the image of a fine line or lines ruled in (or rather cut through) a silver film deposited on glass is formed at the common focus of an object-glass and eyepiece of a telescope.
It should be mentioned that an essential feature of the travelling wire micrometer is that the eyepiece as well as the wire shall be moved by the micrometer-screw.
The scales of the images formed in the focus of the eyepiece common to both microscopes shall be identical.
If the stellar spectrograph is viewed in the focus of 0 2 and the converging rays are reflected by the prism P2 to P4, no image would be seen in the eyepiece, for the rays would pass out directly through the parallel glass plate which is formed by the cementing together of the prisms P 3 and P 4.
Then the lines of the stellar spectrogram would be seen in focus of the eyepiece and the image of the solar spectrograph would be obliterated.
The effect of turning the pinion V is, of course, to displace the focus both of the solar and stellar spectrographs in the field of the eyepiece, but this d .a displacement is easily restored by From Zeitschr.
In polarized light they show a weak grey colour with a black cross, the arms of which are parallel to the cobwebs in the eyepiece of the microscope and remain stationary when the section is rotated.
He therefore selected for any particular measurement such a Rochon prism as when fixed between the eye and the eyepiece (i.e.
Instead of a mirror, the objective of a microscope is attached to one prong of the first fork and the eyepiece of the microscope is fixed behind the fork.
One prong of the fork carries a microscope objective, wart of a vibration microscope, of which the eyepiece is fixed at the back of the clock and the Lissajous figure FIG.
They determined to reinvestigate the motion of y Draconis; the telescope, constructed by George Graham (1675-1751), a celebrated instrument-maker, was affixed to a vertical chimneystack, in such manner as to permit a small oscillation of the eyepiece, the amount of which, i.e.
At the same time, on land, the new necessities imposed on field artillery by the growing use of covered positions led to the development of scissors-telescopes (see Rangefinders) and panorama-telescopic sights (see Sights), in which the optical system was arranged with the tube of the telescope vertical and the object-glass and eyepiece systems at right angles to the axis of the tube.
Immediately above this plane surface and almost touching it is a system of wires which enables angular distances from the centre of the field to be read at the eyepiece below.
By moving the lens G up and down the image can be formed in the correct position for the eyepiece at all extensions of the mast.
(Low power from 3 to 8; medium from 5 to 14; high from 7 to 21.) Each eyepiece is provided with a dummy eyepiece which comes opposite to the eye which is not observing and permits of it being kept open.
The top is closed by a pressure-tight window, inside of which is a prism which reflects the light rays vertically down the tube to a prism at the bottom end, where they are reflected in a horizontal direction and focussed in an eyepiece attached to the bottom of the tube.
In its simplest form it consists of a direct-vision spectroscope, having an adjustable slit (called "camera slit"), instead of an eyepiece, in the focal plane of the observing telescope.
A cross-hair, in the focal plane of an eyepiece, is then moved horizontally until it coincides with the line in question.
The eyepiece is removed and the photographic plate (k) placed in position.